A rapid, accurate, and simple method was implemented for the total chromatographic purity analysis of Chloroquine Phosphate by High Performance Liquid Chromatography with a Diode Array Detector. The experimental conditions follow guidelines, with minor modifications, from the USP43-NF38 monograph methods for Chloroquine Phosphate Assay and Impurity Profiling. Chloroquine Phosphate, Chloroquine Related Compound A, Chloroquine Related Compound D, Chloroquine Related Compound E, Chloroquine Related Compound G, Hydroxychloroquine, and Phenol can be resolved with baseline separation within 16 minutes using an Ascentis Express C18 column (250 x 4.6 mm, 5 µM). A 1.4 g/L Dibasic sodium phosphate solution (pH 3.0) in water: 0.4% triethylamine in methanol (30:70 v/v) were employed as the mobile phase for the isocratic elution. Under applied conditions, system suitability criteria are met, and the method demonstrates good resolution/selectivity, reproducibility, and sensitivity.
Chloroquine Related Compound A
Chloroquine Related Compound D
Chloroquine Related Compound E
Chloroquine Related Compound G
System Suitability Solution (SST) (Standard)
Chloroquine phosphate (Test solution)
Research. Development. Production.
We are a leading supplier to the global Life Science industry with solutions and services for research, biotechnology development and production, and pharmaceutical drug therapy development and production.