Optimization of qPCR conditions is important for the development of a robust assay. Indications of poor optimization are a lack of reproducibility between replicates as well as inefficient and insensitive assays. The two main approaches are optimization of primer concentration and/or annealing temperatures.
The most common application for qPCR is the measurement of a gene transcript or copy number quantity relative to one or more reference genes using probe detection. The reactions may be designed such that each target is detected as a single reaction (simplex) or such that all probes can be added to a single reaction to detect multiple targets simultaneously (multiplex). It is essential that multiplex assays are designed to be compatible with each other. When using probes for SNP or mutation analysis both probes, i.e. specific to mutant and wild type targets are added to the reaction along with a single set of primers.
After optimization of the assays for both target and reference/control genes, these are used to measure the quantity of target. A ratio is determined between the quantity of gene of interest (GOI) and the reference gene.
In this protocol, the probe is included at 250 nM and the primers at 450 nM. However, it is advisable to optimize these conditions and adjust the reaction volumes accordingly. For a multiplex reaction, add all primers and probes to the reaction mix. It may be necessary to use a more concentrated stock for oligos or prepare a concentrated oligo blend.
Notes for this Protocol
Note: Adjust water volume to accommodate addition of additional primers and probes for multiplex reactions or add blended, concentrated stocks.
2. Prepare a 1:10-fold dilution (or suitable dilution) of suitable standard curve template so that there is 20 μL of each dilution
(six dilutions in total).
3. Add 5 μL of appropriate template serial dilution (standard curve), template, controls template or water (NTC) to the
defined tubes or wells.
4. Add 20 μL of master mix to each tube/well. 5. Cap tubes or seal the PCR plate and label. (Make sure the labeling does
not obscure the instrument excitation/detection light path.)
6. Run samples according to Table 2 ensuring that data are collected for all probe labels.
7. See Data Analysis.
Research. Development. Production.
We are a leading supplier to the global Life Science industry with solutions and services for research, biotechnology development and production, and pharmaceutical drug therapy development and production.