Not suitable for clinical diagnostic use. Will not detect clinically relevant species such as M. pneumoniae and U. urealyticum
pkg of 1 kit
dry at room temperature
Optimized for use with JumpStart™ Taq DNA Polymerase, D9307.
NONH for all modes of transport
If available for a given product, the recommended re-test date or the expiration date can be found on the Certificate of Analysis.
The assay kits have a similar range of species detected (see product insert for listing). For Product No. MP0025, the components are supplied as individual reagents. For Product No. MP0035, the components are supplied in reaction tubes, already mixed and ready for addition of sample and Taq. The internal control for Product No. MP0035 is at 481 bp rather than 191 bp as found in Product No. MP0025.
The kit contains 8 positive control reaction tubes and 24 test reaction tubes. A positive control and negative control should be run with each assay. Depending upon the number of samples run with each assay, the kit could test up to 24 samples.
Detection of all mycoplasma species tested so far and usually encountered as contaminants in cell cultures This kit will detect A. laidlawii, M. agalactiae, M. arginini , M. arthritidis, M. bovis, M. cloacale, M. falconis, M. faucium, M. fermentans, M. hominis, M. hyorhinis, M. hyosynoviae, M. opalescens, M. orale, M. primatum, M. pulmonis, M. salivarium, M. spermatophilum, M. timone. This kit is not suitable for detection of M. pneumoniae, U. urealyticum, or other clinically associated species.
Detection of Mycoplasma sp. requires less than 5 mycoplasma genomes per microliter of sample. This is approximately 1-5 fg of DNA.
When a customer observes a band with the positive control and no band for the negative control, it is possible that the Taq could be the problem. We observe a lack of sensitivity with some polymerases from other manufacturers. The activity of the polymerases was enough to amplify the positive control but not the internal amplification control. The internal amplification control is the meter for the sensitivity of the PCR and the activity of the Taq used. The positive control contains more DNA and is good for result evaluation, only. We recommend that the customer use Jump Start Taq (Product No. D9307) as indicated in the protocol.
The kit has been optimized for use with Product No. D9307. Any DNA polymerase with a 3’-5’ exonuclease activity is NOT recommended. These include Tli, Pfu,and Pwo. Those DNA polymerases that may be used include Taq, Tfl, and Tth.
The lot specific COA document can be found by entering the lot number above under the "Documents" section.
There are several ways to find pricing and availability for our products. Once you log onto our website, you will find the price and availability displayed on the product detail page. You can contact any of our Customer Sales and Service offices to receive a quote. USA customers: 1-800-325-3010 or view local office numbers.
Transportation information can be found in Section 14 of the product's (M)SDS.To access the shipping information for this material, use the link on the product detail page for the product.
Research. Development. Production.
We are a leading supplier to the global Life Science industry with solutions and services for research, biotechnology development and production, and pharmaceutical drug therapy development and production.